A. Flow Meter accuracy is guaranteed for a fully developed flow profile in the pipeline at the installed location. Normally from flow disturbance point, flow is developed after a distance of about 20D or more. Hence minimum upstream straight length should be 20D for normal pipelines.
Minimum downstream straight length should be 5D from the flow meter to avoid disturbances in the flow.